High speed – up to 15,rpm/21, x g * Whisper quiet brushless induction drive * Precise, microprocessor control * Automatic g-force conversion program. Hermle ZM-2 Microcentrifuge from Labnet International, Inc. A high performance microlitre centrifuge, offering a fullrange of features:digital display of time, rpm and RCFtwo acceleration and deceleration.
|Published (Last):||19 February 2010|
|PDF File Size:||17.55 Mb|
|ePub File Size:||16.4 Mb|
|Price:||Free* [*Free Regsitration Required]|
Description – The rotor is not installed securely on the motor shaft. Page 20 Error no.: Turn the speed knob 1see figure 4 till the required rotor code appears hermmle the d i s p l a y.
Hermle z233 m-2 Instruction Manual
If the green “slow”control lamp is lit, the unit accelerates and decelerates very softly below 2 rpm. Enter text from picture: The temperature rise depends on the rotor swing-out or angle rotor ,bucket type, ambient temperature, running time and the speed of the rotor.
Operation Operation Operation Operation Operation 4. To print the manual completely, please, download it. Should this happen, the jermle, rotor, buckets, adapters and the rotor chamber must be thoroughly cleaned and all broken particles removed immediately. Got it, continue to print.
If this is not done, they could scratch the protective coating of hemrle rotor. Page 19 Error no.: Width 28 cm Depth 37 cm Height There is a distinction between two different kinds of errors. On the digital display “time min-sec” the letters “ro” rotor appears.
Keyboard – Starting The Centrifuge – “quick”-key 4.
Preselection Of Operating Time 4. Don’t have an account?
HERMLE Z M-2 INSTRUCTION MANUAL Pdf Download.
Page of 20 Go. General Information The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations.
The rotor must always be balanced in this fashion prior to operation. Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap The digital indications on the display will light up when the power is on.
Use only original spare parts! Table of Contents Add to my manuals Add. The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations. Set Or Change Rotor-code -Load the rotor symmetrically by inserting tubes into spaces opposite from each other.
Power failure while rotor is in motion Action: Don’t show me mm-2 message again. Service And Inspection Of Centrifuge 7. During a run and with the centrifuge lid closed you have to press the “preset” key 1hold it and change operating hermls by turning the knob 3.
Figure 8 If the green “fast” control lamp is lit, the unit accelerates and decelerates fast. Commercially available tubes or bottles made of glass or plastic. To open the centrifuge lid.
Comments to this Manuals Your Name. Overcurrent or undervoltage due to power supply fluctuations Action