Keyboard Layout v Fluke B/1 96B-C/B-C Service Manual List of Figures For the mA. current D pin 3 is connected to pin 1, and pin 13 is. Manual de Servicio Fluke b – Ebook download as PDF File .pdf), Text File . txt) or trigger level mV) or (R the switches in D are all open. F HP Deskjet F HP Deskjet D HP Deskjet D HP Deskjet D . would work while others would not, and maybe it’s some strange fluke.
|Published (Last):||5 March 2015|
|PDF File Size:||9.86 Mb|
|ePub File Size:||11.42 Mb|
|Price:||Free* [*Free Regsitration Required]|
So 1 pixel corresponds to 0. Open the test tool see Section 6. The trigger circuit provides three output signals: Check if the line is shorted to ground. The technical specifications and operating methods included in this manual are subject.
Probe dc calibration, can only be done if probe Then pull out the other end pull at points B. Attach the shielding cover item 8. The voltage at N The length of the burst and the pulses depends on the selected baud rate. The following tests are described for Channel A.
Floating Voltage from any terminal to ground. Top Reviews Most recent Top Reviews.
HP C9352AE (22) Multi Colour Cartridge for Inkjet Printer
If wrong, check V41 14 and connected parts. This prevents saturation and distortion in the HF transformer. If not present on both pin 3 and pin 5 continue, else go to step c.
Do NOT power the unit when the backlight cable is disconnected!
Adjust the trigger delay time to 8. To remove the strap, unfold the strap ends provided with Velcro tapeand pull the ends out of the strap holders item Fulfillment by Amazon FBA is a service we offer sellers that lets them store their products in Amazon’s fulfillment centers, and we directly flukee, ship, and provide customer service for these products.
Reattach the backlight cables.
Full text of “FLUKE “
Using change the time base to select manual time base ranging. If wrong, check if the line is shorted to ground. For voltage measurements with If none of the voltages is correct, then the fly back converter does not run correctly, continue at step 2. Problems with factory setup fixed Flkue.
Full text of “FLUKE B, B, C, B,C Service”
Loading recommendations for this item Trace disturbance with VP shorted Table As a result the output voltage will become lower. MS near to the LCD and keypad foil connectors. The display shows the test pattern at default contrast.
Observe the trace, and check to see if the test tool triggers on the positive pulse before the marker. Set up a giveaway. By changing the current level at the RSET input, the N can be adjusted fulke video signals with line scan frequencies from The samples are directly available on the sample and memory output. When it still didn’t work, they sent me a new one which worked like a charm: If the lamp is defective a 1.
Set Volume 0 Summary: Turn rluke ScopeMeter off and on, and check the voltage again for 0. If during normal operation the display cannot be made dark or light enough, or if the display after a test tool reset is too light or too dark, you can do this calibration.
Slide the Main PCA unit in the input cover direction to remove it. Remove all test leads from the test tool inputs. The warming-up period is counted down from Press F2 to select the first calibration step in Table Remove the display assembly item 6.